Configure IP Address -Shark 100
Configuring the Ethernet IP Address requires the following process:
see Doc# E145704 V.1.08
Ethernet Programming (INP10 Option)
Communication Wiring: The meter with Ethernet communication has an RJ45 port on its backplate. Use an Ethernet cable to connect the meter to your PC.
Factory Default Settings: The Factory Default IP parameters programmed in the INP10 card are:
IP Address: 10.0.0.1
Subnet Mask: 255.255.255.0
Set up the PC’s Ethernet Adapter:
The PC’s Ethernet Adapter must be set up for point-to-point communication.
- Select Start>Control Panel>Network Connections. Right-click on the Local Area Connection you will use to communicate with the meter and select Properties.
- Select Internet Protocol [TCP/IP] and click the Properties button.
- Click Use the Following IP Address radio button and enter: IP Address:10.0.0.2; Subnet Mask: 255.255.255.0.
- Click the OK button.